The TransLite Veinlite PEDI2 is a hand-held battery-powered transillumination device for visualizing veins and superficial blood vessels in pediatric patients.
The PEDI2 utilizes the patented1 transillumination technique known as side-transillumination and a patented2 orange color LED light. A ring of bright light from 10 orange and 4 red LEDs is aimed through a clear plastic LED Window  and focused below the skin surface and towards the ring’s center. The focused light creates a volume of illumination, with the central focus acting as a virtual light source under the skin. This technique achieves uniform illumination of a small region of tissue anywhere on the body.
The patented orange light is selected because of high absorption in venus blood. The red LEDs have high penetration in darkly pigmented skins. The combination of the two provides optimum imaging of veins in humans.
|Number of LEDs||10 Orange
|Weight with battery||29g|
|Dimensions||40.8 × 86.6 × 21 mm|
|Battery||2x AAA 1.5V|
Depressing the Power Button  turns on the orange LEDs. Depressing the button again, adds the red LEDs to the orange LEDs. Depressing the button again turn the unit off.
Depressing the Brightness Button  for one second will turn the unit off.
The LEDs will automatically shut off after 7 minutes.
The Veinlite PEDI2 has three levels of brightness that are selected by pressing the Brightness Button  after the unit is turned on. Brightness control is important when working with small children.
DO NOT SHINE THE LED LIGHTS IN THE PATIENT’S EYES.
The single-use disposable plastic covers for the Veinlite PEDI2 (Product #VLP2-DPC) MUST be used when the device is placed in contact with a patient’s skin.
CAUTION: These plastic covers are not sterile. They should be wiped with a germicidal wipe prior to use.
Veinlite PEDI2 is designed for external examination only.
To avoid contamination of the device, Veinlite PEDI2 must be used with the disposable plastic cover.
To avoid the spread of infection, always change the disposable plastic cover between patients and wipe the new cover with a germicidal wipe.
These devices are intended for use only by clinicians trained in performing vascular access, or under the supervision of a clinician.
Do not look directly into LEDs while on.
Device may be wiped down with a germicidal wipe.
Do not immerse the device in liquid.
Do not autoclave the device
Contact Translite about adverse event in relation to this device. Should adverse event occur in E.U., notify EU Competent Authority.
The Veinlite PEDI2 is powered by 2 AAA batteries. They provide between 2 and 4 hours of continuous illumination depending on the number of LEDs turned on and the brightness level used.
When the batteries get low, the orange lights will flicker when turned on. Replace the batteries as soon as the orange light flickers when turned on.
To install or replace the batteries, press firmly on the Battery Cover  and slide it off. Remove any used batteries. Install 2 new AAA batteries, oriented as indicated by the markings inside the battery compartment then slide the battery cover back into place.
The Veinlite PEDI2 has a one-year parts and labor warranty. It is not necessary to register your Veinlite PEDI2 for warranty. The process is automatic, and the serial number of the unit provides all the information needed.
Before returning the device to TransLite for repair, please call or e-mail to obtain an authorization number and shipping details.
No refunds will be given for devices shipped without authorization.