The Veinlite NEO is a hand-held battery powered transillumination device for visualizing veins and arteries in neonates.
To assist in venous and arterial access in neonatal pediatric patients.
|Number of LEDs
|Weight with battery
|96.1 x 19 x 16.5 mm
Depress the Side Button  to change the brightness. Each depression of the button will cycle through low, medium, and high brightness levels. The lowest level is the default setting.
Depress the End Button  each time to cycle through these modes:
|Optimal wavelength for contrast between the venous blood vessels and surrounding tissue.
|Assist in finding arteries and veins.
DO NOT SHINE THE LED LIGHTS IN THE PATIENT’S EYES.
Note: Fragility of preterm infant epidermis is due to incomplete maturation of the skin barrier. Therefore, the Veinlite NEO should be used with care. The VNEO has been designed with a 4-minute auto shutoff and should not be used on the same epidermis location if further use of the device is required.
The NEO should always be used with a single-use disposable plastic cover to protect from contamination and prevent spread of infection between patients.
CAUTION: These plastic covers are not sterile. They should be wiped with a germicidal wipe prior to use.
The device is shipped with 50 covers. Additional covers can be ordered from TransLite or authorized dealers.
Cleaning is not required when disposable covers are properly used. Germicidal wipes, used sparingly, may be used to clean device if needed. Do not submerge the unit in liquid.
Veinlite NEO is designed for external examination only.
To avoid contamination, the Veinlite NEO must be used with the disposable plastic cover.
To avoid spread of infection, always change disposable plastic cover between patients and wipe the new cover with a recommended germicidal wipe.
Always turn the LEDs on after the device is in position.
Do not look directly at the LEDs when they are illuminated or shine directly into anyone’s eyes.
Do not immerse the device in liquid or autoclave
Contact Translite about adverse event in relation to this device. Should adverse event occur in E.U., notify EU Competent Authority.
The NEO is powered by a single AAA battery.
The purpose of the Protective Cap is to protect the End Button , Side Button , and LED Window . Keep a disposable plastic cover on device to ensure a proper fit.
Attach lanyard to Cap .
The Protective Cap and lanyard should be cleaned regularly with alcohol or germicidal solution.
The Veinlite NEO has a one-year parts & labor warranty.
Call or e-mail to obtain an authorization number and shipping details prior to returning device.
No refunds will be given for devices shipped without authorization.
If one of the LED fails to turn on, replace battery. If the problem persists contact TransLite for assistance.
Replace battery, try turning unit on again. If device remains off, contact TransLite for assistance.