The Veinlite EMS PRO® is a hand-held battery powered illumination device for finding veins and assisting in vein access. It has two modes of operation: surface illumination using 8 bright white LEDs and side-transillumination using 12 orange plus 4 red LEDs. The Veinlite EMS PRO uniformly illuminates a small region of skin and subcutaneous tissue, anywhere on the body. The design of the Veinlite EMS PRO is such that during vein access, the vein is visualized, skin is stretched and the vein is secured to keep it from rolling.
|Number of LEDs||12 Orange
|Weight with battery||108.2g|
|Dimensions||115 x 55 x 21 mm|
|Battery||2x AA Alkaline|
DO NOT SHINE THE LED LIGHTS
IN THE PATIENT’S EYES.
Single-use disposable plastic covers (Product #VEMS-DPC) for the EMS PRO MUST be used when devise is placed in contact with patient’s skin.
CAUTION: These plastic covers are not sterile. They should be wiped with a germicidal wipe prior to use.
Veinlite EMS PRO is designed for external examination only.
To avoid contamination, the EMS PRO must be used with disposable plastic covers.
To avoid spread of infection, always change the disposable plastic cover between patients and wipe the new cover with germicidal wipes.
EMS PRO is intended for use only by clinicians trained in performing vascular access, or under the supervision of a clinician.
Do not look directly into LEDs while on.
Contact Translite about adverse event in relation to this device. Should adverse event occur in E.U., notify EU Competent Authority.
Two patented* attachments are included with the EMS PRO. The diagrams below indicate how to attach the light shield and pediatric adapter.
Snap-on light shield to reduce interference from bright overhead lighting.
Snap-on pediatric adapter to reduce size of viewing area.
The EMS PRO is powered by two AA alkaline batteries. Batteries provide between 3-5 hours of continuous use depending on setting and type of battery used.
When batteries run low, an indicator light will be seen under button . A red blinking light indicates approximately one hour of battery life.
To install or replace batteries, press firmly on the battery cap  and slide off.
Remove any used batteries. Install two new AA batteries, oriented as indicated by the markings inside battery compartment. Slide battery cover back into place.
The Veinlite EMS PRO has a one-year parts and labor warranty. Please provide serial number, located under battery or outside of box, when you contact Translite for any warranty issues. The warranty is valid only if the Veinlite EMS PRO is used according to the manufacturer’s instructions. Before returning for repair, you must call or e-mail TransLite to obtain a return authorization code and shipping information before sending your device. No refunds will be given for unauthorized shipping.